Discussion specific to Fine Offset and similar rebadged weather stations
2 posts • Page 1 of 1
- Posts: 498
- Joined: Mon Dec 20, 2010 12:12 pm
- Weather Station: WS-1081 with rain gauge mod
- Operating System: Windows 7 & 1.9.3 b1059
- Location: Palmerston, NT, Australia
foxy1305 wrote:I have noticed over the last few days it appears my wind speed is reading 0. The cups are turning no problem, just not registering inside. Any ideas?
broken cable, cable not plugged in, bad connection (wiggle it), extension cables with bad connections (cut them off and solder together), problem with anemometer (take down and assess the magnet and reed switch), cable not connected to/broken/bad connection with wind vane. Hope that helps.
Michael, Palmerston, NT Australia www.Palmerston-WeatherNo image? I'm offline!
Who is online
Users browsing this forum: No registered users and 4 guests